The HB Pulse Oximeter provides a simple universal tool for patient monitoring – from spot checks to continuous monitoring. All monitoring data can be. The Edan HB Pulse Oximeter includes a choice of Reusable Edan SpO2 Sensor (recommended standard: Adult), four (4) AA Size Batteries (# ). Product Description. The HB employs EDAN SpO2 technology. Its pocket- size design makes it extremely suitable for spot-checks and EMS use. Compact & .
|Published (Last):||9 February 2016|
|PDF File Size:||16.3 Mb|
|ePub File Size:||4.58 Mb|
|Price:||Free* [*Free Regsitration Required]|
The safety and maintenance check can be conducted by personnel from this company. Disinfecting Clean the pulse oximeter before disinfecting it. Electromagnetic propagation is affected by absorption and reflection from structures, objects and people.
HB Pulse Oximeter User Manual interruptions, it UT for 5s is recommend that the product be powered from uninterruptible power supply or a battery. To print the manual completely, please, download it. If there is an error during the POST, the following error codes will display on the screen Do not immerse or wet the sensor, as this may damage the sensor. Do not use the If the value of SpO2 Lo Alarm is set to less than 85, it will restore to 85 after the oximeter is turned on again.
High level alarms Indicate that the patient is in a life threatening situation and an emergency treatment is demanded.
The system consists of Central Processing Unit, Signal Adultfour 4 AA Size Batteries In menu state or trend state, if there is no operation for 30 seconds, the oximeter will return to measurement state.
There are no user-serviceable parts inside the oximeter, the cover should only be removed by qualified service personnel. If you do not hear the POST pass tone, it indicates the alarm system does not work well. The user shall understand that nothing in this manual grants him, expressly or implicitly, any right or license to use any of the intellectual properties of EDAN.
You can set the Sensitivity to Hi or Low via the menu. It can perform spot and continuous measurement for a short time. NOTE2 These oximeger may not apply in all situations.
Edan HB Pulse Oximeter | eBay
About This Manual It is reminded that the product shall be used strictly complying with this manual. Figure Data Transfer State 3.
It’s dedicated to offer reliable, accurate, and sensitive measurement of blood oxygen saturation under even harsh conditions, such as in hospitals, intra-hospital transport, and other hospital grade facilities. Page 4 EDAN may define as user serviceable.
Pulse Oximeter | EDAN H100B
Don’t have an account? If the operating time of a Ni-MH rechargeable See the appropriate sections of the manual for specific safety information. This equipment is not intended for family usage. The recommended disinfectants include: Backlight control and automatic power-off function for power saving.
The sensor unconnected icon and associated alarm indicate the sensor has disconnected or wire fault. Green indicates that the charging is complete. The latest 10 minutes trend graph and table of SpO 2 and Pulse Rate can be reviewed in the screen.
Items that are certified or that are listed as seller refurbished or used have been cleaned and certified to proper operating specifications, and are in excellent condition. Page 21 HB Pulse Oximeter User Manual The symbol indicates that the device should be sent to the special agencies according to local regulations for separate collection after its useful life. Failure to cover the sensor site with opaque material in high ambient light conditions may result in inaccurate measurements.
Notes are identified by the symbol shown above. HB Pulse Oximeter User Manual maximum output power rating of the transmitter in watts W according to the transmitter manufacturer and d is the recommended separation distance in metres m.
Wash the probe with clean water after disinfecting it to remove any remaining solution. Battery Charger Stand Kit. Red and infrared low-voltage light-emitting diodes LED serve as light sources; a photonic diode serves as the photo detector. Table of Contents 1 Safety Information If the measured field strength in the location in which the oximeter is used exceeds the applicable RF compliance level above, the oximeter should be observed to verify normal operation.
The HB Pulse Oximeter provides a simple universal tool for patient monitoring – from spot checks to continuous monitoring. HB Pulse Oximeter readings and pulse signals can be affected by certain ambient environmental conditions, sensor application error, and certain patient conditions.
Page 3 The user shall understand that nothing in this manual grants him, expressly or implicitly, any right or license to use any of the intellectual properties of EDAN. Table of Contents Add to my manuals Add. Audible and visible alarm capability adjustable.
This device has been tested and found to If service is necessary, contact qualified service personnel or your local EDAN representative.
Authorized representative in the European community Date of manufacture Manufacturer Part Number – 12 Figure Confirm to Cover Data 3. Law restricts this device to sale by or on the order of a physician. Page 90 HB Pulse Oximeter User Manual interruptions, it UT for 5s j100b recommend that the product be powered from uninterruptible power supply or a battery.
Food and Drug Administration and state and local regulatory agencies.