D CV. Read this owner’s manual thoroughly before use Owner’s operation manual for series. DT Series. 3 1/2 Digital Multimeter. Read this owner’s manual thoroughly before use. DT series multimeter have been designed according to IEC concerning electronic measuring. View and Download Histest DT owner’s operation manual online. DT Multimeter pdf manual download.
|Published (Last):||23 October 2009|
|PDF File Size:||7.22 Mb|
|ePub File Size:||13.76 Mb|
|Price:||Free* [*Free Regsitration Required]|
Histest DT830 Owner’s Operation Manual
This function allows the meter to output a signal as a signal generator. Table of Contents Add to my manuals Add.
Close case and tighten screws completely before using the meter to avoid electrical shock hazard. If the defect has been caused by a misuse or abnormal operation conditions, the repair will be billed manula a nominal cost. Page of 13 Go. Related Manuals for histest DT No related manuals. Set the range switch to the desired range. A test signal will appear between the and “COM” jacks.
Disconnect circuit power and discharge all high-voltage capacitors before testing resistance, continuity, diodes, or capacitance. The voltage is approximate 5Vp-p. Before attempting to open the case, manuap be sure that test leads have been disconnected from measurement circuits.
By pressing ‘print’ button you will print only current page. In the latter case, the transient stresses are variable; for that reason, its necessary that the transient-withstand -capability of equipment is made known to the user. Got it, continue to print. Before measuring current, check the meter’s mannual and turn power off to the circuit before connecting the meter to the circuit.
Set the range switch to ” ” or ” ” range. Front Panel Description This function allows the meter to output a signal as a signal generator. Don’t have an account? Battery And Fuse Replacement Before attempting to open the case, always be sure that test leads have been disconnected from measurement circuits.
Maual the proper terminals, function, and range for your measurements.
DT-830D Manual Ranging Digital Multimeter
Enter text from picture: If the resistor to be measured is connected to a circuit, disconnect circuit power and discharge all capacitors before measuring resistance. To print the manual completely, please, download it. Page 6 To use safely, each measurement can not last for more than 10 seconds, and the interval between each measurement must be more than 15 minutes.
Operating Instruction “COM” jack. Don’t show me this message again.
DTD Manual Ranging Digital Multimeter
Comments to this Manuals Your Name. A coupling capacitor should be used when connecting the meter to a circuit.
Safety Information If the defect has been caused by a misuse or abnormal operation conditions, the repair will be billed at a nominal cost. To use safely, each measurement can not last for more than 10 seconds, and the interval between each measurement must be more than 15 minutes.